Optec® 5000/5000P Vision Screeners
Ideal for pre-screening, to quickly and accurately test patients in a controlled environment.
Perform comprehensive visual screening in minimal time. Screen under a variety of conditions, including monocular, binocular, near, distance, and intermediate. The Optec 5000P can also function as a peripheral vision test. Adult and pediatric visual acuity testing, color and depth perception, as well as vertical and lateral phorias.
The Optec 5000/5000P with Manual Control includes:
- Accessory Kit*
- One standard slide package
- The 5000P includes the peripheral test
*Accessory kit includes: dust cover, pointer, lens cleaner, headrest tissues, recording forms, and instruction manual
Clean and disinfect the forehead activator and surrounding area gently with disinfectant wipes or a soft, slightly damp, and lint-free cloth with 70% isopropyl alcohol. Clean gently and never spray alcohol directly on the screener surfaces.
- Quick, repeatable and accurate
- Lightweight and portable
- Distance/Near Lens Systems
- See and point observation doors on both sides
- No training or certification needed
- Includes a slide package with a set of 12 test slides as well as the option to create a custom vision testing slides library from more than 150 test slides
STANDARD TEST PACKAGES HAVE BEEN DESIGNED FOR:
- Public Health
- Primary Care
- Occupational Medicine
- Driver Rehabilitation
- Also Customizable Test Packages
STANDARD SLIDE PACKAGES FOR OPTEC 5000 SERIES:
- Ophthalmic Prescreening
- Michigan Preschool
- Driver Rehabilitation
- Patti Pic
The package of 12 slides may be customized at the time of purchase at no additional cost. However, no F.A.C.T. contrast sensitivity slides may be included in these standard packages.
How do I clean and disinfect the screener cover?
Clean and disinfect the forehead activator bar and surrounding area gently with disinfectant wipes or a soft, slightly damp, and lint-free cloth with alcohol. We have used 70% isopropyl alcohol to clean and disinfect our equipment surfaces. Clean gently with a damp cloth and never spray alcohol directly on the screener surfaces.
Why am I not getting ‘light’ to my instrument?
Both left and right eye buttons (at least one) must be illuminated on the control panel for the instrument to receive light.
Why are some of my slides up side down?
Test slides will appear upside-down or reversed when looking at a near point slide at a far point setting (and vise versa).
What lighting condition is acceptable for the testing room?
Any lighting condition is acceptable for day testing. Night testing should be administered in a dimly lit room.
How do you change the light bulbs in the vision tester?
There are no light bulbs in the vision tester. The unit is run solely on LED’s (Light Emitting Diodes).
How to Fix an Upside Down Slide
How to Change a Slide
Headrest Tissues – Optec 5000, 2000 & 1000 Series (5000/pkg)